We keep getting errors 410 and 420 from the switch. If the connection error is 34970, then it is likely network related and you should make sure the Service Location Provider component of Novell has been installed and configured for your There can only be one Runtime User credential for DNS stored in CASA.If you want to run novell-named as any non-root user other than named, you need to set the credentials Bookmark Email Document Printer Friendly Favorite Rating: novell-named will not resolve newly added records even after a restartThis document (7003447) is provided subject to the disclaimer at the end of this http://edvinfo.com/login-error/login-error-code-8804.html
I'm helping a colleague with an application measuring ten channels each of temperature, frequency, and current using an Agilent 34970A DAQ switch (GPIB) and LabVIEW 7. Logged USMCPMEL 7-level Posts: 799 Action Taken: +7/-13 Re: Agilent 34970A error codes « Reply #5 on: 03-16-2009 -- 08:48:07 » Ok anyone know where this mystery battery might be? Got that figured out now I have to figure out how to reset them all. So basically a good bit of the calibration constants have somehow been wiped out on this unit.
Show 6 replies Re: Hi all, I have agilent 34970A, and when I turn it on, a repetitive error message code 305 appears , I don't know how can I fix michaelkellett Jan 25, 2015 10:57 AM (in response to samir6044) This is a current Agilent product. Looking at the user manual, it appears that the error is related to calibration memory of the unit. original site The Service Manual states it's a subset of the complete list.
Document ID:7003447Creation Date:04-JUN-09Modified Date:07-MAY-12SUSESUSE Linux Enterprise Server Did this document solve your problem? please if someone have solution thanks. Or you may havetried to configure scaling on a module that does not connect to theinternal DMM.MK Like Show 1 Likes(1) Reply Actions Re: Hi all, I have agilent 34970A, and Product Alias/Synonym enrole tim itim isim Document information More support for: IBM Security Identity Manager Adapters Software version: 5.1, 6.0, 7.0 Operating system(s): Windows Software edition: All Editions Reference #: 1380551
Everything worked good I got this thing reading spot on except for 4 wire ohms. http://www.ibm.com/support/docview.wss?uid=swg21380551 Logged MRD 3-level Posts: 95 Action Taken: +2/-0 Re: Agilent 34970A error codes « Reply #8 on: 03-16-2009 -- 11:40:26 » Did you look in the Users Manual, 34970-90003, for the The system returned: (22) Invalid argument The remote host or network may be down. ICP 备案号 10220084.Premier Farnell plc, registered in England and Wales (no 00876412), registered office: Farnell House, Forge Lane, Leeds LS12 2NEelement14 Software Version: 188.8.131.52 , revision: 20160519114300.21bdc7e.stable_8.0.4.x United States English English
I looked up the codes and they are all cal data lost codes. EEVblog Electronics Community Forum A Free & Open Forum For Electronics Enthusiasts & Professionals Welcome, Guest. I'm at home now, I'll check some of my information at work tomorrow and see if there's anything else I can find out. http://edvinfo.com/login-error/login-error-ensure-you-have-the-correct-login-information.html with error 25May 21 11:08:42 fs01 [NCPL]: zone 0.22.10.IN-ADDR.ARPA/IN: loaded serial 2009041421May 21 11:08:42 fs01 [NCPL]: Unable to login Error code:34970May 21 11:08:42 fs01 [NCPL]: Unable to get Journal Log attribute
In the case described by this technote, everything is specified correctly, yet the connection can still not be made. DisclaimerThis Support Knowledgebase provides a valuable tool for NetIQ/Novell/SUSE customers and parties interested in our products and solutions to acquire information, ideas and learn from one another. John Treekiller Logged msingh Pinger Posts: 2 Action Taken: +0/-0 Re: Agilent 34970a Error Codes « Reply #4 on: 07-22-2010 -- 10:03:58 » Thanks for the help, I will try this
All Places > Manufacturers > KEYSIGHT TECHNOLOGIES > Discussions Please enter a title. First i go the errormessage 205 on the Agilent instrument. 225: Settings conflict: DMM disabled or missing This command is valid only when the internal DMM is installed and enabled. Like Show 0 Likes(0) Reply Actions Re: Hi all, I have agilent 34970A, and when I turn it on, a repetitive error message code 305 appears , I don't know how michaelkellett Jan 26, 2015 8:03 AM (in response to samir6044) Is the instrument is connected to anything else ?Have you done the power on self test ?Have you cleared the configuration.Does
The vendor did a self test diagnosis and the problem went away. I use 19200 baud, 8-bit, no parity for the COM settings. The data appears to be valid, but due to a network problem, the client cannot determine how to connect to the server. 34948 is returned by the Novell Client to say Check This Out How to fix? (Read 2249 times) 0 Members and 2 Guests are viewing this topic.
Query Unterminated happens when the instrument is addressed to talk but there is no command pending that will place data in the buffer. Like Show 0 Likes(0) Reply Actions Go to original post Hi all, I have agilent 34970A, and when I turn it on, a repetitive error message code 305 appears , I You mayhave tried to a configure a channel for current measurements (validonly on channels 21 and 22 on the 34901A module). I have an automated adjustment procedure for this.
samir6044 Jan 25, 2015 5:22 PM (in response to michaelkellett) Thanks michaelkellett , I read the error message but what I can do ? All Rights Reserved. If the errors go away then the fram is indeed dead.A lot of these machines are flogged on ebay with statement 'does not power up' or 'cal errors'. plazma Regular Contributor Posts: 141 Country: HP/Agilent 34970A with error codes.
Watson Product Search Search None of the above, continue with my search Novell Adapter connection error novell 34970 connect connection multicast SLP Technote (troubleshooting) Problem(Abstract) Attempting to test the Novell adapter The driver is pretty good but that doesn't mean that there isn't a bug or two that will show up when the instrument is being used in a manner that wasn't Please login or register.Did you miss your activation email? 10-20-2016 -- 01:21:44 1 Hour 1 Day 1 Week 1 Month Forever Login with username, password and session length Top 10 Posters Figure.
Please try the request again. That's the multimeter module.Is that the full list or are there others ? This is the "Fault Tolerance" mode designed to let DNS load and function using the last good information retrieved from eDir...--- if there are no /etc/opt/novell/named/*db files then the daemon will Logged Print Pages:  « previous next » PMEL Forum» Forum» General» General Discussion» Agilent 34970a Error Codes DISCLAIMER: This site is not an official US Air Force site, it